There are two different methods (wireless and wired) for programming the Phonak Audéo M depending on the model. On Page 6 in the this link (Phonak Audéo Marvel Product information PDF) the following two methods are specified;
- All models can be programmed wirelessly via Noahlink Wireless
- The larger (in size) models with Telecoil and/or size 13 battery can be programmed with a wired programming device, namely the mini pro or Hi Pro
For both methods (wired and wireless programming) you will need version 6.0 (in some cases 6.0.1) or higher of Phonak Target fitting software.
Wireless Programming via Noahlink Wireless:
The Noahlink Wireless device is connected to your computer via USB;
For wired programming using a wired programming device (mini Pro or Hi Pro) you will need Phonak/Unitron CS44a cables that have pins 3&4 reversed from the standard CS44 cables.
mini Pro with USB cable to connect to your computer
CS44a cables look the same as this standard CS44 cable except pins 3&4 are reversed
Closeup of the cable-end that connects to the hearing aid, if you look closely you can find visual alignment markings (red dot) and physical alignment (grove that only fits one way).