Hi Guys , thought I should share you my experience and let me know your opinion.
I use Phonak Naida V30Up in both ears
I also have V50 UP as a back up.
For mobile and streaming I use Compilot Air II.
I do my own programming,and have tried all the permutations and combinations till I get the right combination that suits me.
I find the following combination works for me:-
1.Ear mould:- Hard, as it reduces chances of ear infection and irritation, and is suitable for long hours usage
- Fitting Software;- DSL V5 adult. The sounds are clearer and sharper. I have tried NAL and Adpative Phonak Digital, but I find DSL more clearer in speech
3.Always choose “Damped”,it provides more power in sound.
4.By default , Occlusion is selected.But I change to Hearing settings to 0.9-1.1mm for left, and right 1.7-2mm , it provides more power and comfort to my hearing.
5.Noiseblock I select to max, this reduces noise
6.Ultrazoom is not comfortable for me, I change to real ear or Fixed directional.
7.MPO I choose to max 137db.
Overall this works for me, and call/music quality via Compilot is crystal clear…
Felt above combination after a lot experimentation should be shared to you.However it may not be suitable for all, it also depends on the range of your loss.
Let me know your opinion…